User:DandreaBarth415

Disposable Infusion Set with needle Product performance and the main structure The product by the intake device, the stopper puncture, water only cards drip bucket, catheters, injection parts, joints of the outer cone, precision filter / liquid filter, precision flow regulator, membrane back to the blood transfusion needle, protective sleeve. In the infusion process, with the sensitivity back to the blood auxiliary judgment. The scope of application The product with the film back to the blood, venipuncture blood back difficulties (such as: obesity, lower blood pressure, vascular conditions, emergency rescue and children) of patients, the product with precision flow regulator with a stable flow role. Of taboos Is not clear. [Note] Check the package expiration date is expired. 2 investigation infusion packaging with or without leakage, puncture and infusion needle protective cover is off, such as off Do not use. 3 NA infusion flow. Flow scale precision flow regulator given are approximate value for reference only, the flow regulator should be subject to the unit time drops drip bucket (see packaging). This product is one meter static head corresponding relational tables, using the flow rate of 0.9% saline test flow, the flow of basic error see "basic error". 6. Flow setting adjustment device does not apply to high-viscosity liquids. 【Warning】 This product tube routing DEHP plasticized polyvinyl chloride (PVC) material, there are more than two decades of history of use, there is no significant damage reported in the literature of such products for human health. Reported in the literature, the large doses of DEHP in some small experimental animals (such as mice, rabbits, dogs, etc.) there is a certain poison. In the clinical use of the process, if the alternatives, to make use of alternatives to high-risk groups (such as infants and young children, before puberty male, pregnant women and lactating women) treatment. Some medications may produce 4.PVC adsorption phenomenon, reducing the infusion volume of the active ingredients. (5) This product should not be stored and infusion of fat emulsion and fat-soluble liquid and drugs. Note that the possible interactions of the products and infusion drugs, prohibited drugs incompatible with PVC and DEHP infusion, such as paclitaxel, in addition to the clinical expressly prohibit the use of this product infusion of drugs, no this product Other adverse drug reactions.